It is located at the upper portion of microscope. The head portion of microscope is also known as the Body tube.
Microscope head application:
The head portion or body tube of microscope connects the eyepiece to the objective lenses.
If you were to open the body of the microscope you would find a mirror or prism depending on the type and quality of the microscope. The prism or mirror is used to reflect the light and reorient the image by making it appear right side up.
In compound microscopes with two eye pieces there are prisms contained in the body that will also split the beam of light to enable you to view the image through both eye pieces.
Binocular zoom microscope head body
Binocular zoom body is the main body of a stereo microscope, with continuouszooming function and observe with two eyepieces
This body tube usually be placed on microscope stand
Magnification 6.7X-45X, 3.4X-270X with optional eyepieces and objectives
Eyepiece WF10X/22mm high-eyepoint, wide-field; Convenient for observers wearing glasses
Optional Eyepiece WF10X/22mm, WF15X/16mm, WF20X/12mm, WF25X/9mm, WF30X/8mm
Main Body Binocular head, 45° inclined, 360° rotatable, can be locked in any position desired
Interpupillary Distance adjustment:54-75mm; Diopter adjustment:±5 diopters
Objective 0.67X-4.5X stereo zoom; Zoom Ratio:6.7:1
Aux Lens 0.3X, 0.5X, 0.7X, 0.75X, 1.5X, 2X